According to EN 196-3, EN 480-2** optional equipment is required.
ConstructionThe 8 measuring points (2 x 4) are installed in a rectangular trough
(Go to ASTM variation or go to EN/ASTM combination)
Tests can be performed in air or in water bath with constant temperature (heat exchanger and cooler for constant water temperature)
Programmed test sequences
Automatic registration of the beginning and of the end of setting time, as well as of setting plots
Weight of plunger and the Vicat needle are an additional 300 g ± 1 g
After each penetration the Vicat needle is automatically cleaned
Scope of deliveryBase unit as a table model with the control unit
Standard software for programmable test sequences and for data recording
8 Vicat moulds (conical hard plastic ring) 70/80 dia. x 40 mm
8 glass base plates, 120 mm dia.
8 centerings for Vicat moulds (conical hard plastic ring) 70/80 dia. x 40 mm
2 pair of cleaning brushes
1 tee wrench, 1.25 mm
1 tee wrench, 2 mm
2 spare glass plates, 120 mm dia.
2 spare initial-set needles, 1.13 mm dia.
Filter sponge coarse
Filter sponge fine
USB / RS232-Adapter
Dimensions(W x D x H) = 700 x 600 x 600 mm
Weight: 62 kg
110 - 230 V / 50 - 60 Hz
Important notes ** For operation is a standard PC or laptop required, which is not included in the scope of delivery
** For standard EN 480-2 the additional weight 1.0366.10, the plunger 1.0366.12 and optionally the extra cooling unit 1.0366.13 for a test temperature of 5°C are required